The 300 Digital Temperature Probe has a 1.5mm thin tip for fast response. With huge digits that is one easy-to-read and easy-to-use thermometer for rapid checks when food preparation a variety of foods.
You are watching: Comark 300 thermometer change to fahrenheit
Digital Temperature Probe
The 300 Digital Temperature Probe has a 1.5mm thin guideline for rapid response. With big digits the 300 is an easy-to-read and also easy-to-use thermometer for rapid checks when cooking a variety of foods.
There is a hold switch on the 300 which permits you to host the digital reading on the screen while you keep in mind a measurement.
The 300 Digital Temperature Probe is additionally water-resistant.Field CalibratableData hold FunctionAccurate to ±1°C / ±2°F1.5mm Thin tip ProbeWater Resistant
The 300 Digital Temperature Probe is a basic reduced (thin) guideline Thermometer v Biocote Antimicrobial Protection. The 300 is slim and also lightweight, which renders it perfect because that slipping into shirt, trouser or apron pockets in between use. It has actually a 127mm / 5″ long stem and a 1.5 mm / 0.6″ diameter climate tip, making it perfect for probing thinner foods. The 300 Digital Temperature Probe is ar calibratable and features °F/°C switchable temperature scales/ranges, a data hold duty and a short battery indicator. You deserve to either usage the ON/OFF switch to conserve battery life or permit the unit automatically shut turn off after a quick time. This bag Digital Thermometer contains a protective instance with pocket clip and a battery.
The 300 is also accessible with a Rubber Boot, click here.Field Calibration Procedures
The 300 is only calibratable in a slush ice systems (60 percent crushed ice cream to 40 percent insanity water) or a water bath. Execute not attempt to calibrate in any type of other manner
Immerse the stem that the thermometer to in ~ least fifty percent its size in the slush ice solution
Wait for the reading to come to be stable (usually ±2° that 32°F or ±1° the 0°C)
Press and also hold the CAL button for 8 secs to begin calibration
CAL will be displayed for two seconds and also calibration at 0°C / 32°F is completed
Remove the thermometer from the ice water and continue to take measurements
Note: CAL will be displayed any type of time the D-H key is pressed for 8 seconds, but the thermometer will not it is in re-calibrated unless the pointer is put in ice cream water°F/°C Switchable
The 300 Digital Temperature Probe functions °F/°C switchable temperature scales/ranges. Through the thermometer turned OFF, press and also hold the ON/OFF button. Release as soon as the desired scale is shown.
See more: How Many Bonds Does O2 Have, How Can Oxygen Have Three Bonds
Note: This is the procedure for any pocket Digital Thermometer that does not have actually an °F/°C scale button.
|Temperature measure up Range||-40°C come +150°C / -40°F come +300°F|
|Scales||°C and °F|
|Accuracy||±1°C / ±2°F|
|Stem Length||127mm / 5"|
|Probe tip Diameter||1.5mm|
|Battery Type||1.5 V switch Cell battery (Part no. AB13; LR44)|
|Conformance come EU Regulation 1935/2004 ("Framework" Regulation on every Food contact Materials)||Yes|
|Stem (Shaft) Material||SUS304 Stainless Steel|
|N/A||Technical Datasheet (300/300B)||02||21st October 2019|
|20534||US Product selection Catalog||02||24th April 2017|
|20536||Thin guideline Thermometers Datasheet||01||24th April 2017|
|EU DoC||EU declaration of Conformity (300)||01||6th November 2014|
|1935/2004||EU Regulation 1935/2004 Compliance||01||21st November 2017|